Surface Roughness Measurement
Gauge Stand SRS-1
Installation & Operation:
Installation Procedure: Unloading the screws in adjustment bracket with a screwdriver, remove the adjustment bracket, mounted the instrument on the measurement stand(use two previously removed screws). Operation Procedure: Press the power button to boot, by adjusting the measuring bracket so that the sensor contact with the measured object, the stylus cursor position (refer to manual component description) should be instructed in “0” point (or upper and lower frame) position. After adjustment, the measurement can be taken in 2 seconds when the instrument is stable.
- Instrumentation and measurement stand should be fixed, to prevent loosening, shaking, causing measurement errors.
- After adjustment, the measurement can be taken when the instrument is stable in 2 seconds, ensure the accuracy of measurement. (the stylus will shake when adjusting position)
- When measuring curved surface, should try to make the measurement distance as short as possible,suggestion sampling length at 0.25mm.